Digital HF Power and SWR Meter RED-DOT 1.6Mhz to 60Mhz 2017A max power 199W
Connect the output of your transceiver to the "TX" and the Antenna or Load to the "ANT".
Please NOTE: This UV-Band meter utilize "so239" type connectors. Proper adaptor / cable is required if your device has different type of connectors.
The latest model 2017A HF VSWR Power Meter adopts sampling techniques, the A/D converter effective resolution arrives to 12bits.It remains easy operation and many other practical features.The RF connector is the most commonly used in HF band called M-KF (also known as SO239 UHF, SL16, etc.)
Battery installation: Remove 4 black round screws on the corner of back cover and open it. Install 3 AAA batteries. BE CAUTIOUS to the polarities. Put the cover on and fasten the screws. IMPROPER INSTALLATION OF BACK COVER WILL CAUSE ERROR ON U-BAND MEASUREMENT!
Connect the Output of your transceiver to "TX" and Antenna or Load to "ANT".
NOTE: UV-Band meter utilize "N" type connectors; and"SL16" connectors (normally named "M" connecter) for HF-Band. Proper adaptor/cable is required if your device has different type of connectors.
Press the red button to power on. After 2 seconds of LCD self test, it's ready for normal measurement.
UV-Band meter will display in sequence:"+ XXX P" (pass through power),"- XXX P" (feed back power)," XXX Γ" (SWR value), 2 seconds display for each mode. Press the red button will hold the current display. Press it again to continue the display.
Under the environment of night or weak illumination, just press and hold the red button for 1 second to turn the blue back light on. Press and hold the red button for 1 second again to turn it off.
8 minutes after no power was measured, it will power off automatically. Press and hold the red button for 2 seconds to turn off manually.
1x Digital VHF/UHF Power & SWR Meter (Battery not included)